Designed especially for hydroponic, aquaponic, and greenhouse growers who need to test pH, EC, TDS and temperature, the Hanna Instruments GroLine HI9814 is a combo meter that measures all four of these parameters.
The GroLine HI9814 is waterproof, easy to use, and features Hanna’s exclusive QuickCal mode. While other meters on the market use two separate probes for pH and EC/TDS, are bulky, and have lower accuracy, the Hanna HI9814 delivers accurate results with one singular probe, is ergonomic, and stabilises and calibrates in about 30 seconds. With its affordable price and practical design, users of any level can professionally monitor pH, EC/TDS and temperature for just $437.80.
Check out the GroLine HI9814 in action now with this handy user guide, and see just how easy we’ve made it to use.
About the GroLine pH/EC/TDS HI9814 Meter
The HI9814 GroLine waterproof meter lets you measure pH, electrical conductivity (EC), temperature, and total dissolved solids (TDS) in parts per million (ppm). It’s the ideal nutrient meter for a range of purposes in agriculture, horticulture, aquaponics, and hydroponics.
The GroLine HI9814 is the most accurate pH/EC/TDS meter in its class and is one of Hanna Instruments’ best selling products.
A major feature of the HI9814 is the amplified electrode. This electrode reduces electrical interference, which is especially important for an accurate reading in a hydroponics environment where equipment like ballasts and lights are running.
The HI9814 also features a Quick Cal function to make calibration fast and simple. Quick Cal can calibrate both pH and EC — a trait that no other meter in the agriculture market has.
Step 1: Rinse the electrode with deionised water to remove any salt deposits that may have formed around the protective cap.
Step 2: Install the batteries into the proper compartment. Be sure to tighten the battery cover completely to ensure water-tightness.
Step 3: Connect the probe by placing it into the socket, aligning the pins. Tighten the nut to ensure a secure fit.
Step 4: Press the ON button to turn on the meter.
Setting Up the HI9814 GroLine Meter
On power up, the meter will display the last selected parameter, temperature, stability indicator, and instructions on how to enter calibration/setup modes.
Selecting the correct EC or TDS unit
Step 1: To enter Setup Mode, press and hold the MODE button for six seconds. OFF and CAL messages will appear before entering Setup Mode.
Step 2: Press the SET button to choose EC UNIT (for conductivity), 500 UNIT (for 0.5 TDS factor), or 700 UNIT (for 0.7 TDS factor).
Hanna Tip: We are often asked when to use 0.5 or 0.7 PPM. You need to match the PPM values based on the values given by your fertiliser manufacturer’s nutrient feed charts. These feed charts don’t always specify which scale their PPM values are based on; in these cases, contact the company directly.
Selecting calibration type
Step 1: After you select your EC/TDS setting, press MODE to go to the CAL setting.
Step 2: Press the SET button to choose either CAL STD for standard calibration, or CAL QUIK for the one-point quick calibration.
Hanna Tip: Use Quick Cal whenever possible. When calibrating using the fast, one-point Quick Calibration feature compared with the two-point Standard Calibration option you will save time and money on calibration solutions. The QuickCal feature works with the HI500036P Calibration Solution Sachets.
Selecting temperature units
Step 1: After you select your calibration setting, press MODE to go to the TEMP setting.
Step 2: Press SET to select either °F or °C.
Selecting auto-off time
Step 1: After you select your temperature setting, press MODE to go to the AOFF setting.
Step 2: Press SET to select an AOFF interval at either 8 minutes, 60 minutes, or to disable (—).
Hanna Tip: The meter can be used as a wall mounted monitor when the AOFF feature is extended to 60 minutes or is disabled. If battery life is a concern, eight-minute auto off is the best option.
After you select your auto-off setting, press MODE to return to the measurement screen.
Now that the meter is set you your liking, let’s calibrate the probe. Calibration of the Hanna HI9814 GroLine Meter is made simple with our Quick Calibration Solution HI50036P.
Quick Calibration Steps
Step 1: Clean your electrode by removing the protective cap and rinsing the probe in distilled or deionised (DI) water. This removes any impurities or salts that may contaminate the calibration solution.
Step 2: Press and hold the MODE button for three seconds until CAL is displayed to enter calibration mode.
Step 3: Immerse the probe in an open packet of quick calibration solution, stirring gently.
Step 4: Wait for stabilisation. Once stable, the meter will automatically calibrate. The display will read CAL OK, then return itself to normal measurement mode.
You are now free to take your measurements!
Hanna Tip: If you happen to be out of the Quick Cal solution then use the Standard Calibration mode with the standard pH 4.01 and pH 7.01 buffers and 1413 µS solution. Make sure that you watch the screen carefully so you don’t miss the prompt to change the buffer or solution! Missing a prompt will result in an error and a failed calibration.
Get your sample ready before performing the following steps. (Scroll down for instructions and a video on the best way to prepare a soil sample.)
Step 1: Before taking a measurement, rinse the probe in distilled or DI water.
Step 2: Place the probe in your sample, and wait for a stable reading. Pressing the SET button will let you switch between pH and EC/TDS readings.
Hanna Tip: Make sure that you stir the sample. Stirring ensures the sample is well mixed and increases stability speed; a light stirring is all it takes. After stirring, ensure that you wait until the reading stabilises. (We often see people taking the reading too early. It’s better to take extra time to make sure that the reading is stable.) Finally, clean the electrode at least weekly or as needed with the Hanna Cleaning Solution for Agriculture Applications HI700661P or general purpose cleaning solution.
Step 1: Once finished with measurements, press the OFF button to power down the meter. Rinse your probe with distilled or DI water to prevent any biological growth on the glass of the electrode.
Step 2: Add a small amount of storage solution to the protective cap, and place the cap over the probe. This ensures that your probe stays hydrated until your next use.
Hanna Tip: Make sure that you store the electrode out of direct light to minimise algae growth. Your pH meter should also be stored in a cool place; letting the meter sit out under the heat of high-powered grow lights will cause premature wear and tear.
It’s important to periodically clean your probe. Setting a maintenance schedule can help you keep on top of it.
Step 1: Place the probe in a packet of Cleaning Solution for Agriculture HI700661P for 15 minutes.
Step 2: Rinse the probe with distilled or DI water.
Step 3: Place the probe in storage solution for at least 2-3 hours.
After cleaning your probe, you will need to recalibrate it before taking any measurements.
Get your Hanna GroLine HI9814 now on our website!
P.S. We’re Making it Even Easier!
The Hanna Conical Tip pH Electrode HI1294D allows you to test your soil or soil medium quickly and directly.
Video: Hanna Conical Tip pH Electrode (1 Minute)
If you have any questions about our GroLine pH/EC/TDS HI9814 Meter, contact us by phone at (03) 9769 0666 or email at firstname.lastname@example.org
Download the Brochure: